[amsat-bb] Re: AMSAT-BB Digest, Vol 3, Issue 619 SINAD Testing
jeffrey.davis at me.com
Sun Nov 30 08:39:01 PST 2008
Hello The Net:
SINAD is nice because the complete path from RF thru IF thru audio
demodulation can be tested.
Also a cost effective approach, Last SINAD meter I got was about 100$.
It is a HELPER Instruments,Inc. model SINADDER LINEAR 5
SINAD measurements require an external signal generator, and variable
A noise figure meter can usually test the RF path or the RF thru IF
path, but NOT to audio.
For my HP-8970B, the IF would have to be more than 10 MHz, the low freq
end of the machine.
NF meters are about 10-20X $$$ the SINAD meter purchase.
I use my NF meter in a Y factor mode, which is essentially a signal to
SINAD is also a Signal to Noise ratio, with Distortion taken into
(S+N+D)/(N+D). When noise is very low, and distortion is also low, thus
SINAD was used by Motorola and FM radio testing with a 1000Hz test
Testing can also be used for testing SSB receivers, using a clean
carrier with no modulation
and tuning the RX for max signal, max S/N, max SINAD reading.
Both instruments have their utility.
Stan, W1LE Cape Cod, FN41sr
> 2. Re: MMIC for LEO preamp? (RFI-EMI-GUY)
Sent via AMSAT-BB at amsat.org. Opinions expressed are those of the author.
Not an AMSAT-NA member? Join now to support the amateur satellite
Subscription settings: http://amsat.org/mailman/listinfo/amsat-bb
More information about the AMSAT-BB